#		<< IRIS SEED Reader, Release 4.5 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     ARMA
B050F16     Network:     AU
B052F03     Location:    ??
B052F04     Channel:     BHZ
B052F22     Start date:  2007,169,00:00:00
B052F23     End date:    2007,171,23:59:59
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),  ARMA ch BHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - velocity in meters per second
B053F06     Response out units lookup:             V - emf in volts
B053F07     A0 normalization factor:               5.71405E+08
B053F08     Normalization frequency:               0.3
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       5
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -7.402000E-02  7.402000E-02  0.000000E+00  0.000000E+00
B053F15-18    1 -7.402000E-02 -7.402000E-02  0.000000E+00  0.000000E+00
B053F15-18    2 -5.027000E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -1.005000E+03  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -1.131000E+03  0.000000E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,  ARMA ch BHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  8.084240E+02
B058F05     Frequency of gain:                     3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,  ARMA ch BHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  4.000000E+00
B058F05     Frequency of gain:                     3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),  ARMA ch BHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - emf in volts
B054F06     Response out units lookup:             COUNTS - digital counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,  ARMA ch BHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     6.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,  ARMA ch BHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  4.993800E+04
B058F05     Frequency of gain:                     3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),  ARMA ch BHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 4
B054F05     Response in units lookup:              COUNTS - digital counts
B054F06     Response out units lookup:             COUNTS - digital counts
B054F07     Number of numerators:                  29
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -3.013550E-04  0.000000E+00
B054F08-09    1 -1.103390E-03  0.000000E+00
B054F08-09    2 -1.343540E-03  0.000000E+00
B054F08-09    3  8.525790E-04  0.000000E+00
B054F08-09    4  5.848280E-03  0.000000E+00
B054F08-09    5  9.972140E-03  0.000000E+00
B054F08-09    6  6.348320E-03  0.000000E+00
B054F08-09    7 -9.171890E-03  0.000000E+00
B054F08-09    8 -3.040360E-02  0.000000E+00
B054F08-09    9 -3.896790E-02  0.000000E+00
B054F08-09   10 -1.327990E-02  0.000000E+00
B054F08-09   11  5.515500E-02  0.000000E+00
B054F08-09   12  1.498740E-01  0.000000E+00
B054F08-09   13  2.332600E-01  0.000000E+00
B054F08-09   14  2.665230E-01  0.000000E+00
B054F08-09   15  2.332600E-01  0.000000E+00
B054F08-09   16  1.498740E-01  0.000000E+00
B054F08-09   17  5.515500E-02  0.000000E+00
B054F08-09   18 -1.327990E-02  0.000000E+00
B054F08-09   19 -3.896790E-02  0.000000E+00
B054F08-09   20 -3.040360E-02  0.000000E+00
B054F08-09   21 -9.171890E-03  0.000000E+00
B054F08-09   22  6.348320E-03  0.000000E+00
B054F08-09   23  9.972140E-03  0.000000E+00
B054F08-09   24  5.848280E-03  0.000000E+00
B054F08-09   25  8.525790E-04  0.000000E+00
B054F08-09   26 -1.343540E-03  0.000000E+00
B054F08-09   27 -1.103390E-03  0.000000E+00
B054F08-09   28 -3.013550E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,  ARMA ch BHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 4
B057F04     Input sample rate:                     6.000000E+02
B057F05     Decimation factor:                     3
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             2.333300E-02
B057F08     Correction applied (seconds):          2.333300E-02
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,  ARMA ch BHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 4
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),  ARMA ch BHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 5
B054F05     Response in units lookup:              COUNTS - digital counts
B054F06     Response out units lookup:             COUNTS - digital counts
B054F07     Number of numerators:                  5
B054F10     Number of denominators:                5
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  1.832160E-04  0.000000E+00
B054F08-09    1  7.328640E-04  0.000000E+00
B054F08-09    2  1.099300E-03  0.000000E+00
B054F08-09    3  7.328640E-04  0.000000E+00
B054F08-09    4  1.832160E-04  0.000000E+00
#		Denominator coefficients:
#		 i, coefficient, error
B054F11-12    0  5.174780E-01  0.000000E+00
B054F11-12    1 -2.409340E+00  0.000000E+00
B054F11-12    2  4.238860E+00  0.000000E+00
B054F11-12    3 -3.344070E+00  0.000000E+00
B054F11-12    4  1.000000E+00  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,  ARMA ch BHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 5
B057F04     Input sample rate:                     2.000000E+02
B057F05     Decimation factor:                     10
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             1.000000E-02
B057F08     Correction applied (seconds):          1.000000E-02
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,  ARMA ch BHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 5
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,  ARMA ch BHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           1.614840E+08
B058F05     Frequency of sensitivity:              3.000000E-01 HZ
B058F06     Number of calibrations:                0
#		
